Surge tester built to the requirements of IEC 60601-1:2005 (Issue 3) Figure 10 (IEC 60601-1 Figure 50 in older versions). Conducts IEC 60601-1:2006 Figure 11..
Surge tester built to the requirements of IEC 60601-1:2005 (Issue 3) Figure 10 (IEC 60601-1 Figure 50 in order versions).
Conducts IEC 60601-1:2006 Figure 11; IEC 60601-2-34 Issue 2 (optional); IEC 60601-2-34 Issue 3 (included);
IEC 60601-2-49; EC-13; EC-53 and the new requirement of IEC60601-2-25:2011 and IEC 60601-2-27:2011 Tests of
protection against the effects of defibrillation (Common mode, Differential mode), Energy reduction test (EC-13). All these
tests use the same 5000V source with difference in output wave shaping. The various wave shaping output circuits required
for each test are included in the D5-PF.
The more remarkable changes f the D5-PF version are: output voltage selected digitally, direct joules readout, increased
accuracy, faster time between pulses "20 seconds", increased signal generator voltage, pass/fail reference is integrated,
dry cap is used and USB Testminder is standard and our unique feature integrated energy measurement.